Este sitio web utiliza cookies para mejorar su experiencia de navegación. Más información sobre cómo usamos las cookies y cómo puede cambiar su configuración.
Alemania-Dresde: Aceleradores de partículas
Anuncio de licitación
Apartado I: Poder adjudicador
Direcciones de internet:
Dirección principal: http://www.hzdr.de
Apartado II: Objeto
The electron linear accelerator ELBE consists of 2 electron injectors and multiple user beamlines that can be partly operated in parallel. An important sub-system of the machine is the timing signal generation and distribution which controls electron bunch emission, beamline diagnostic and user experiments on the picosecond timing level. The system needs to be capable for continuous operation which demands a technology readiness level of TRL-8 (“System completed and qualified through test and demonstration”). This call aims for the development, integration and commissioning of a timing system at the ELBE center for high-power radiation sources. The system should contain of a central timing master unit which has interfaces to the ELBE control system and machine protection system. The ELBE control system is based on Siemens PLCs and uses WinCC v7.3 for visualization.
The timing master has to generate flexible timing signals that are variable in frequency, duty cycle, delay and number of cycles. Logical combinations of timing signals should be used to derive advanced trigger output signals. Minimum 10 receiver stations need to be connected to the timing master via electrically isolated links. The receiver stations must generate user defined pulse patterns on a scalable number of physical outputs (electrical and optical) based on the timing master configuration. The system must be maintainable and extendable by the machine staff in terms of number of receiver stations, physical outputs, timing signals and their configuration.
— Minimum Hardware requirement:
— max. trigger rate: 26 MHz,
— delay step size: 200 ps,
— pulse duration step size: 200 ps,
— min. delay range / minimum signal period: 300 s,
— max. trigger jitter [1 Hz, half trigger frequency] for best performing output type: < 30 ps RMS,
— reference clock frequency: 1300 SRF-Accelator; 13 MHz master oscillator,
— resynchronization frequency (parameter set/update, counter/divider reset): 10 Hz (Draco), 1 Hz (Penelope),
— different Output Logic Level: TTL, (LV)PECL,
— minimum number of timing client stations: 10,
— minimum number of different independent output signals per Client: 10,
— minimum software requirement.
All the sources of the development that are needed to maintain, extend or modify the system, in particular hardware design files, firmware sources and software program code, must be provided to the purchaser. The development must be documented in an appropriate manner for the different system components like program code comments, block diagrams, component lists and functional descriptions.
Used Hardware components as spare part.
Apartado III: Información de carácter jurídico, económico, financiero y técnico
Declaration about the registration within a professional or trade register as prescribed by the legal provisions of the state where the bidder is based.
Declaration regarding the exclusion criteria specified in Sections 123, 124 GWB.
Declaration regarding the total turnover of the company and turnover with respect to the special type of task, the subject-matter of the award procedure, relating in each case to the last 3 business years.
Minimum sales of special task type per year ≥ 1 000 000 EUR
Specification of work performed in the last 5 years.
≥ 3 References comparable with the tendered task in their nature and scope. Each reference have to include all the following specifications:
a) the application should be in the field of particle accelerators (Linac, Synchrotron, Medical accelerator and similar);
b) the number of client receiver stations should be at least 8;
c) the reference implementation has high demands on the availability of the timing system (24/7 operation) and is operated routinely; and
d) must not be a demonstrator setup.
Apartado IV: Procedimiento
Apartado VI: Información complementaria